We are known for manufacturing and exporting of Arty Forceps in Sialkot, Punjab, Pakistan.
Cracks in the Box Lock: Cracks in the box lock develop due to continuous stress on the instrument and are often unavoidable. These small cracks significantly reduce the structure integrity of the instrument and usually occur in one of three places indicated below. Cracks should be apparent if looked at closely, but can often be missed without proper inspection. Instrument with a cracked box lock usually can be repaired and should be sent in immediately.
Jaw Alignment: One of the most common reasons a forceps needs repair is misalignment of the jaws. For a standard forceps or hemostat inspecting the jaws with the instrument securely closed should reveal if it is improperly aligned. For forceps with teeth, open and close the instrument to feel if the teeth are snagging rubbing each other.
Loose Box Lock: A loose box lock is another common problem which occurs from normal wear and tear. To check if the box lock is to loose open the instrument, grab one ring handle in each hand and gently push one handle up and down. There should be some play in the instrument, but if it feels a sloppya it should be repaired.
Levior impex, Established in 2010 at Sialkot in Punjab, is leading Exporter, Manufacturer of Surgical Instruments in Pakistan. Levior impex is one of Trade India's verified and trusted sellers of listed products. With their extensive experience of supplying and trading Needle Holder Forceps, Levior impex has made a reputed name for themselves in the market with high quality Needle Holder Forceps, TC Webster Needle Holder, Surgical Scissors etc. Focusing on a customer centric approach, Levior impex has a pan-India presence and caters to a huge consumer base throughout the country. Buy Surgical Instruments in bulk from Levior impex at Trade India quality-assured products.
- BUSINESS TYPE Exporter, Manufacturer
- ESTABLISHMENT 2010