We are instrumental in offering our customers an excellent quality range of Wrist Strap Tester. Wrist strap is the first line of defence against electrostatic charges builds up on the human body. It is designed to carry away this charges as soon as it is generated. At the same time wrist strap must remain electrically safe for the user. Thus, it is important to test every wrist strap regularly.
In order to protect the sensitive electronic components the maximum wrist strap resistance should not be more than 10 Meg ohms on the other end. The resistance should not be less than 0.75 Meg ohms as protection to the wearer from dangerous level of the voltage and current flow.
Our range of Wrist Strap Testers checks these parameters under actual operating conditions with the strap on the wrist. Wrist Strap Tester clearly indicates weather or not the resistance of the wrist strap system including the wearer is at least 0.75 Meg ohms and not more than 10 Meg ohms respectively.
Range : Wrist strap 0.75 to 10 Meg ohm,
Indications : Pass â€“ Green Led
Low & High Fail â€“ Red Led With Audible alarm
Test actuation : By gently pressing metal sensor plate,
Accuracy : +/- 10%,
Power Supply : 9 Volt Battery
Low Battery : Warning at 7.0 V,
Dimensions : 135 x 70 x 30 mm
Weight : App. 150 Gms with battery,
Calibration : Recommended every 12 months
Warranty : 12 Months,
Optional : Instruction Board, Carry Case & Door Locking System.